CADET
Key Features

Stationary or removable breadboard area

Multiwaveform function generator

Variable output AC/DC power supply

8 selectable logic switches

8 High and Low buffered logic indicators

2 SPDT logic switches

1k and 10k potentiometers

2 debounced pushbutton operated pulsers
 2 BNC's and 8 ohm speaker
Specifications
 POWER
 3wire AC line input 230V 50Hz. Illuminated power switch.
 POWER SUPPLIES:
 Fixed 5VDC @ 1.0A, ripple <5mV
 Variable +1.3V  15V 0.5A, ripple <5mV
 Fixed AC O/P 12.6VAC Centre tapped @ 100mA
 FUNCTION GENERATOR:
 Frequency Range: 0.1Hz to 100kHz in six ranges
 Output voltage: 0 to +10V (20V pp)
 Output impedance: 600 ohms (except TTL)
 Output current: 10mA max., short circuit protected
 Output waveforms: Sine, Square, Triangle, TTL
 Sine wave distortion: <3% (10Hz to 100kHz)
 TTL pulse: Rise and fall time <25ns, Drive: 10 TTL loads
 Square wave: Rise and fall time <1.5us
 LOGIC INDICATORS:
 16 LED's; 8 red to indicate logic high, 8 green to indicate logic low
 DEBOUNCED PUSHBUTTONS (PULSER):
 Two pushbutton operated, opencollector output pulsers, each with one normallyopen and one normallyclosed output.
 Each output can sink upto 250mA.
 POTENTIOMETERS:
 One 1k ohm, one 10k ohm, all leads available and uncommitted.
 SPDT SWITCHES:
 Two SPDT slide switches, all leads available and uncommitted.
 BNC CONNECTORS:
 Two BNC connectors, pin available and uncommitted, shell connected to ground.
 LOGIC SWITCHES:
 8 individual SPDT logic switches, one side of all eight switches connected to 0V, the other side of all eight switches is switchable between +5V and +V.
 Wipers are uncommitted and are taken to tie points.
 SPEAKER:
 0.25W, 8 ohm.
 BREADBOARDING AREA:
 A total of 2520 uncommitted tiepoints. Two bus strips internally connected to power and ground.
 Fifty tie points for each +5V, +V, V and ground.
 ENCLOSURE:
 Plastics moulded case.
 DIMENSIONS:
 165mm(H) x 410mm(W) x 290mm(D), 3.2kg
No comments:
Post a Comment